The MAC 1200 is a compact 12-lead electrocardiograph designed for acquisition - with or without analysis. Remote learning daysmrs. parker's 6th grade l.a. class. The system can be configured as a basic electrocardiograph without analysis for facilities with minimal ECG requirements. The system can also be configured to incorporate the proven 12SLTM ECG analysis program.
Adding the 12SL program creates an electrocardiograph with the industry leading interpretation program and measurements. If your facility only requires measurements, we offer a version with 12SL measurements only.
3 months ago
|3 months agoTime and Date issue|
The coin cell is at 3.25 VDC. The hardware test states that system memory and static memory both errored out. Is this a coin cell issue or the main board? Any assistance would be greatly appreciated.Reply
7 months ago
|7 months agonot printing in auto mode|
i can not print in auto mode . what i could be the problem ?Reply
9 months ago
|9 months agoFlashing LCD Upon Startup|
When I have the unit plugged in to a AC, the green standby light is on and when I press the 'on/ stdby' button the green LED by 'start' and yellow LED by 'battery low' and 'stop' will flash intermittently.I have tried a new main battery in the unit and that does not help.The power output from the power supply to the PCB is 25.8 DCV. Would not having the called for 26.5 DCV cause this issue or is that close enough to the called for output?Reply